Bill– showing us the guidewire and dilator coming together…
This is a three part series- “IABP 101” designed to simply refresh or introduce the basic concepts of applying diastolic augmentation for the high risk cardiac patient. Again, some of the concepts here will seem to be fairly rudimentary for a seasoned perfusionist, but if it’s your first day in a perfusion program- then elementary reverts to elemental- as in a prerequisite foundation for any training program delivered to perfusionists.
This section details preperation and insertion of the IABP- more information is available at the manufacturer’s website:
A shout-out to Bill and Kim– who are in the lead role at this institution (in terms of managing the IABP’s as well as in-servicing the CCL staff). Nicely done!
Have an excellent day!
IABP: Parts & Pieces
Guide Wires INTRODUCER
Guide Wires IABP
USE ONLY for Sheathless Insertion
IABP- Sizing, Placement, and Insertion
Sheath or Sheathless?
Sizing The IABP
Preparing the IABP
Aspirate 30cc to ensure IABP is collapsed- DO NOT remove one-way valve!
Remove stylette from inner lumen: DO NOT replace stylette
Place stopcock on luer port and flush with 3-5 cc NSS. DO NOT allow air to enter!
Use the Blue T Handle as a marker for proper positioning- usually between the 2nd and 3rd intercostal space…
The balloon should be located in the proximal descending aorta, just below the origin of the left subclavian artery. This ideally results in the balloon terminating just above the splanchnic vessels 3.
Initiating the IABP
If you have power issues- check this 1st!
Hook up IABP extension tubing to helium outlet
Attach Fiberoptic cable: DO NOT touch the end with your fingers!
Press START to auto fill, zero, and self time the IABP
IABP timing itself…
Attach Slave cable to back of IABP
Zeroing Pressure When Slaving to Ext Monitor
This is the same concept as “opening to air” when zeroing regular transducers
This must be done within 15 seconds of pressing the VENT button